This instrument is produced based on ASTM D1895 Standard Test Methods for Apparent Density, Bulk Factor, and Pourability of Plastic Materials.
It is mainly used for the determination of the apparent density of loose materials flowing out from the specified funnel. It is an ideal test instrument for the quality inspection of raw materials entering the chemical industry.
2.Structure and operation method
(1)Structure: The analyzer consists of a base, a support frame, a funnel, and a receiver.
a.Open the package, take out the analyzer, and check the parts according to the packing list.
b.Place the analyzer on a stable work surface.
c.Block the lower outlet of the stainless steel funnel with any flat material, and place the receiver under the funnel, coaxial with the funnel.
d.Take a sample of 410ML-420ML from the measuring cylinder and slowly place it in the funnel.
e.Quickly remove the insert, so that all the samples are naturally leaked (the receiver that is filled with the sample must not be shaken).
f.Use a ruler to scrape off the excess sample on the mouth of the receiver, and then weigh it on the drug balance to the nearest 0.2g (the sample must not be repeatedly measured).